Background: Glaucoma is a group of diseases characterised by retinal ganglion cell dysfunction and death. Detection of glaucoma and its progression are based on identification of abnormalities or changes in the optic nerve head (ONH) or the retinal nerve fibre layer (RNFL), either functional or structural. This review will focus on the identification of structural abnormalities in the RNFL associated with glaucoma.
Discussion: A variety of new techniques have been created and developed to move beyond photography, which generally requires subjective interpretation, to quantitative retinal imaging to measure RNFL loss. Scanning laser polarimetry uses polarised light to measure the RNFL birefringence to estimate tissue thickness. Optical coherence tomography (OCT) uses low-coherence light to create high-resolution tomographic images of the retina from backscattered light in order to measure the tissue thickness of the retinal layers and intraretinal structures. Segmentation algorithms are used to measure the thickness of the retinal nerve fibre layer directly from the OCT images. In addition to these clinically available technologies, new techniques are in the research stages. Polarisation-sensitive OCT has been developed that combines the strengths of scanning laser polarimetry with those of OCT. Ultra-fast techniques for OCT have been created for research devices. The continued utilisation of imaging devices into the clinic is refining glaucoma assessment. In the past 20 years glaucoma has gone from a disease diagnosed and followed using highly subjective techniques to one measured quantitatively and increasingly objectively.
Statistics from Altmetric.com
Funding: Supported in part by National Institutes of Health contracts R01-EY13178-09, R01-EY11289-23 and P30-EY08098-20 (Bethesda, Maryland), The Eye and Ear Foundation (Pittsburgh, Pennsylvania) and unrestricted grants from Research to Prevent Blindness, Inc. (New York).
Competing interests: JSS receives royalties for intellectual property licensed by Massachusetts Institute of Technology to Carl Zeiss Meditec. GW received research funding from Carl Zeiss Meditec and Optovue. JSS received honoraria from Carl Zeiss Meditec, Heidelberg Engineering and Optovue. KAT reports no conflicts.
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.