Aim To compare the cube and radial scan patterns of the spectral domain optical coherence tomography (SD-OCT) for quantifying the Bruch’s membrane opening minimum rim width (BMO-MRW).
Methods Sixty healthy eyes and 189 glaucomatous eyes were included. The optic nerve head cube and radial pattern scans were acquired using Spectralis SD-OCT. BMO-MRWs were automatically delineated using the San Diego Automated Layer Segmentation Algorithm. The BMO-MRW diagnostic accuracy for glaucoma detection and rates of change derived from the two scan patterns were compared.
Results There was a significant difference between the baseline global BMO-MRW measurements of cube and radial scans for healthy (301.9±57.8 µm and 334.7±61.8 µm, respectively, p<0.003) and glaucoma eyes (181.2±63.0 µm and 210.2±67.2 µm, respectively, p<0.001). The area under the receiver operating characteristic curve for differentiating between healthy and glaucoma eyes was 0.90 for both the radial scan-based and cube scan-based BMO-MRW. No significant difference in the rate of BMO-MRW change (mean follow-up years) by scan pattern was found among both healthy (cube: −1.47 µm/year, radial: −1.53 µm/year; p=0.48) (1.6 years) and glaucoma eyes (cube: −2.37 µm/year, radial: −2.28 µm/year; p=0.45) (2.6 years).
Conclusion Although the cube scan-based BMO-MRW was significantly smaller than the radial scan-based BMO-MRW, we found no significant difference between the two scan patterns for detecting glaucoma, identifying BMO location and measuring the rate of BMO-MRW change. These results suggest that although BMO-MRW estimates are not interchangeable, both scan patterns can be used for monitoring BMO-MRW changes over time.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.