Aim: To evaluate secondary glaucoma after penetrating keratoplasty with anterior-segment optical coherence tomography (OCT).
Design: Case series.
Methods: Four eyes of four patients with corneal opacity and increased intraocular pressure (IOP) were evaluated using high-speed (2000 axial scans/s) OCT at 1.3 μm wavelength. Cross-sectional images of the anterior segment were analysed to assess the cause of increase in pressure.
Results: Slit-lamp evaluation of the anterior chamber in all cases was limited by corneal opacity. The OCT imaging allowed visualisation of anterior-segment structures behind the opaque corneas. Using OCT, iris–intraocular lens adhesion and pupillary block were identified as the probable reasons for the increased IOP in one case. Peripheral anterior synechiae and angle closure were identified in the three remaining cases. In two cases, we found that the tip of the aqueous drainage tube was blocked by peripheral anterior synechiae.
Conclusions: OCT is similar to ultrasound in that it allows visualisation through opaque corneas. However, OCT has an advantage in that it requires neither contact nor immersion. It is a valuable tool for evaluating the depth of the anterior chamber angle and the causes of secondary angle closure.
- IOL, intraocular lens
- IOP, intraocular pressure
- OCT, optical coherence tomography
- PAS, peripheral anterior synechiae
- UBM, ultrasound biomicroscopy
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
Published Online First 13 September 2006
NIH R24 EY 13015 grant from National Eye Institute, Bethesda, Maryland, USA; research equipment and grant from Carl Zeiss Meditec, Dublin, California, USA; an unrestricted grant from Research to Prevent Blindness, New York, New York, USA.
Competing interests: DH receives royalty from OCT patents licensed to Carl Zeiss Meditec, Dublin, California, USA. D H and Y L receive research grant support from Carl Zeiss Meditec.