Statistics from Altmetric.com
Endothelial cell density (ECD) assessment with the noncontact Topcon SP2000P specular microscope is known to be reliable when the automated mode is followed by manual corrections (touched-up mode). We compared its agreement with Rhine-Tec, a new noncontact specular microscope, in 270 eyes of 160 patients, by comparing the ECD measured in the automated and touched-up modes with that of Topcon touched-up. Good agreement existed between either touched-up modes with a mean difference of only 2 cells/mm2 95% CI (−27; 23) whereas agreement with the Rhine-Tec automated mode was poor with an overestimation by a mean of 226 cells/mm2 95% CI (172; 281).
The Topcon SP2000P non-contact specular microscope (Topcon, Tokyo, Japan) is widely used to measure corneal endothelial cell density (ECD). It uses a cell contour recognition algorithm based on contrast differences, and ECD derived from an automated delineation of cell boundaries with manual correction of inaccurately drawn cells (“touched-up” mode) has been validated.1,2 A new commercially available non-contact specular microscope Rhine-Tec (Rhine-Tec, Krefeld, Germany) determines ECD by a cell-centre method …
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.