Background/aims: A limited number of scans compromise conventional optical coherence tomography (OCT) to track chorioretinal disease in its full extension. Failures in edge-detection algorithms falsify the results of retinal mapping even further. High-definition-OCT (HD-OCT) is based on raster scanning and was used to visualise the localisation and volume of intra- and sub-pigment-epithelial (RPE) changes in fibrovascular pigment epithelial detachments (fPED). Two different scanning patterns were evaluated.
Methods: 22 eyes with fPED were imaged using a frequency-domain, high-speed prototype of the Cirrus™ HD-OCT. The axial resolution was 6 μm, and the scanning speed was 25 kA scans/s. Two different scanning patterns covering an area of 6×6 mm in the macular retina were compared. Three-dimensional topographic reconstructions and volume calculations were performed using MATLAB™-based automatic segmentation software.
Results: Detailed information about layer-specific distribution of fluid accumulation and volumetric measurements can be obtained for retinal- and sub-RPE volumes. Both raster scans show a high correlation (p<0.01; R2>0.89) of measured values, that is PED volume/area, retinal volume and mean retinal thickness. Quality control of the automatic segmentation revealed reasonable results in over 90% of the examinations.
Conclusion: Automatic segmentation allows for detailed quantitative and topographic analysis of the RPE and the overlying retina. In fPED, the 128×512 scanning-pattern shows mild advantages when compared with the 256×256 scan. Together with the ability for automatic segmentation, HD-OCT clearly improves the clinical monitoring of chorioretinal disease by adding relevant new parameters. HD-OCT is likely capable of enhancing the understanding of pathophysiology and benefits of treatment for current anti-CNV strategies in future.
Statistics from Altmetric.com
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.
Competing interests: Carl Zeiss Meditec provided an HD-OCT-System to support the scientific collaboration. Further commercial relations: None.
Ethics approval: The study complied with the declaration of Helsinki and was approved by the local ethics committee.